An updated version of the Millivolt Meter project from May 2016, with dual range selection, calibration selection, improved software, and updated board layout.Continue reading “Millivolt Meter Version 2”
Louis at Scullcom Hobby Electronics has a new video updating the milliohm meter, and features the board I produced in this post.
NEW Link to updated video on YouTube
NEW Version 1.5 board available at OSH Park requires additional 0.01 uF ceramic capacitor, 2 pin header, and 250 mA PTC fuse.
NEW Updated project post with Milliohm Meter version 1.5 board.
It uses a 4 wire kelvin measurement system that supplies a precise 100 mA current to the device under test using one pair of leads, and then measures the voltage drop across the device under test using a second pair of leads.
As designed it can be used to measure low resistances of under 2 ohms (Ω). Some of the design changes I made to fit my needs were:
- Change the number of resistors needed in the current source circuit
- Use higher precision resistors with a low temperature coefficient
- Add separation and shielding between input, output, and supply circuits
- Use lower ESR and higher value capacitors for the MAX680 power supply
- Add additional decoupling capacitors on the INA106, and MAX680 IC’s
- Change the voltage regulator from a TO220 package to a SOT-223 package